TekBox TBL5016-1 50µH LISN AC/DC CISPR 16
The TekBox TBL5016-1 50µH LISN is a device required to setup conducted noise measurements of DC or AC-powered electronic equipment. It is designed according to CISPR 16-1-2 and MIL-STD-461F standard and characterized in the frequency range of 9kHz to 100 MHz.
TekBox TBL5016-1 LISN AC/DC maximum current: 16A continuous.
Apart from carrying out conducted noise measurements, LISNs are also used to establish a defined impedance for various other standardized tests.
The LISN is a 50Ω/50µH+5 Ω single line design with a high current jumper to short the 5 Ohm resistor and configure it as 50Ω/50µH LISN according to CISPR 16-1-2. The LISN is characterized up to 100 MHz which means that the LISN can be used for the combined range of 9 kHz to 100 MHz.
The LISN is inserted into the supply line of the DUT (Device Under Test). Conducted noise, which is present at the supply terminals of the DUT can be measured at the BNC connector using a spectrum analyzer or a measurement receiver. The source (supply) terminal and the DUT terminal are decoupled by a 50µH inductor. DC or single phase measurements typically require a pair of TBL5016-1. Combining it with the Tekbox LISN Mate enables separate measurement of common mode and differential mode noise. DUTs with 3 phase supply require three or four pieces of TBL5016-1, depending on whether the product is supplied in delta or star configuration.
The unit comes equipped with a BNC-male to N-male RG232 cable, mating connectors, high current configuration jumpers and ground brackets to attach it to a ground plane.
LISN type: V-AMN, configurable as 50Ω/50µH+5 Ω or 50Ω/50µH
Frequency range: 9 kHz – 100 MHz
DC Resistance: < 45 mΩ
Maximum current: 16A continuous
Operating voltage range: 0 – 250V DC; 0 – 250V AC (50/60 Hz), 0 – 120V AC (400 Hz)
Fuse: 2 x 16A slow
High current plug/screw terminals – male: Phoenix Contact 1998933, female: Phoenix Contact 1967375
Dimensions: 250 mm x 225 mm x 140 mm (including ground brackets)
weight: 2.7 kg
I ordered a desoldering station that got lost in transit but they asked regularly for updates and sent it to me again, so they solved the problem in a timely manner.
Client on 30/06/2020
Volevo ringraziarvi per la competenza,serietà nel vostro settore e la cortesia del personale chiamato! Ho ricevuto il materiale nei tempi previsti e in perfetto stato. Ancora da accendere e provare...
Massimo on 30/06/2020