TekBox TBMDA3 modulated wideband power amplifier
The TekBox TBMDA3 modulated wideband power amplifier is designed in order to create an inexpensive signal source for immunity testing of electronic building blocks and products. It is designed to be driven by the tracking generator output of spectrum analyzers. With its high gain, it can boost the output power of any tracking generator up to 5W. The TBMDA3 is ideal to drive Tekbox near field probes in order to find the sensitive spot of an electronic circuit or to create electric fields up to 550V/m when driving the Tekbox TEM Cell TBTC0, 300V/m when driving the TBTC1, 150V/m when driving the TBTC2 or 100V/m when driving the TBTC3. Test signals for immunity testing can be CW, AM or PM modulated. Consequently, the TBMDA3 provides built in modulation capability to generate 1 kHz AM or PM signals. In PM mode, the TBMDA3 can also generate a 217 Hz Signal with 12.5% duty cycle in order to simulate mobile phone TDMA noise.
CW amplifier (modulation off)
1 kHz, 80% AM modulation
1 kHz, 50% duty cycle pulse modulation
217 Hz, 12.5% duty cycle pulse modulation
Input / Output: 50 Ohm, N female
Supply Voltage range: 110 V…240 V
Supply power consumption: 20 W
Maximum input power: +3 dBm
Operating temperature range: -20°C to 50°C
Frequency range: 10 MHz – 1 GHz
1dB output compression point @ 10MHz: +36 dBm typ
1dB output compression point @ 100 MHz: +37 dBm typ
1dB output compression point @ 500 MHz: +36 dBm typ
1dB output compression point @ 1 GHz: +34 dBm typ
Harmonics: < – 10 dBc typ.
Internal modulation frequency AM: 1 kHz ±10%
Internal modulation frequencies PM: 1 kHz ±10%, 217 Hz ±20%
Duty cycle, PM: 50% ±10% @ 1 kHz; 12.5% ±20% @ 217 Hz
general purpose power amplifier
signal source for immunity testing, driving near field probes
signal source for immunity testing, driving TEM Cells
I ordered a desoldering station that got lost in transit but they asked regularly for updates and sent it to me again, so they solved the problem in a timely manner.
Client on 30/06/2020
Volevo ringraziarvi per la competenza,serietà nel vostro settore e la cortesia del personale chiamato! Ho ricevuto il materiale nei tempi previsti e in perfetto stato. Ancora da accendere e provare...
Massimo on 30/06/2020