KIT 5 TG EMI/EMC analizzatore 7.5GHz con TG + LISN + PC SW
KIT 10 EMI/EMC analizzatore di spettro 9kHz - 7.5GHz con Tracking Generator + LISN CISPR 16 + software per PC + accessori
- Rigol DSA875-TG analizzatore di spettro 9kHz - 7.5GHz con Tracking Generator
- opzione EMI- Filter & Quasi-Peak Detector kit
- opzione Pre Amplificatore
- SW per PC Rigol EMI
- NFP-3 RF Near Field Probe Kit
- TekBox TBLC08 LISN - CISPR 16
- cavo di collegamento BNC-BNC tra LISN e DSA
- adattatore RF-N to BNC
EMI Test System is a PC application software developed by RIGOL for DSA1000A, DSA1000 and DSA800 (with the EMI- DSA800 option) with the EMI function. This software is designed on the basis of the standard drive VISA and you can realize the communication between the software and instrument via USB-TMC or LAN interface to control the instrument.
Users can perform conduction and radiation tests using EMI Test System and RIGOL DSA series spectrum analyzer. You can measure the interference voltage on the power cable using the linear impedance stability network (LISN) and perform amplitude correction on the results by loading the correction factor (antenna, cable, other or user) automatically in the radiation test.
This software also provides various functions to facilitate your measurements. You can set various parameters (such as the frequency range, resolution bandwidth and scan time) via the scan list. After performing a scan, the results can be displayed in log or linear format. You can search for signal, measure its peak value, quasi-peak value and average as well as display the results in the peak list. You can mark and delete the undesired signal as well as easily recognize signals that do not pass the standard limit line by using the peak list function.
- Provide amplitude correction function.
- You can edit the scan list and perform scan by segments to improve the measurement speed.
- The limit line function can be used to quickly judge the measurement results.
- Provide fast pre-scan and final scan modes.
- Provide peak search function. You can define and save the peak list.
- The frequency axis supports linear and log scale display.
- Auto generation of test report.
For users to use the software to quickly perform EMI test, the measurement procedures as shown in the figure below are recommended.
Edit the amplitude correction data
Enter amplitude correction setting and compensate the gain or loss of the external devices (such as the antenna and cable). You can view the correction data list in table form as well as save and load the correction data currently edited.
Set the scan parameters
Configure the pre-scan, segment scan and final scan parameters. You can set the segment scan parameters separately using EMI Test System, view the segment scan data list in table form as well as save and load the segment scan data currently edited.
Edit the limit line
Edit the limit line data and set the measurement limits. You can view the limit line data list in table form as well as save and load the limit line data currently edited. After performing a step frequency scan, you can preview the measurement results in the spectrum window and compare them with the preset limit line value.
Perform segment pre-scan according to the segment scan setting to improve the measurement speed of the software.
Set the peak parameters and perform peak search. The spectrum analyzer filters and marks the peak list according to the user- defined conditions. Users can edit the peak list, add or delete frequency points as well as save and load the peak list currently edited.
One-key final scan. You can perform more accurate scan on the critical interference signal using the final scan function to ensure the measurement accuracy of the software.
Add report parameters and other explanations according to the actual test environment. The software generates a test report (*.html file) automatically for further processing of the measurement values.
Purchase at 12 and shipped at 15. Package received 3 days after . Batterfly was very reactiv and efficient . On the next day it was in use for our project . Very satisfied . I will recommend it
Laurent on 29/09/2020
very good product, fast delivery and excellent services
Floricica Alexandru on 28/09/2020